- Automatic measurement and adjustment of cuff pressure
- Reduced risk of ventilator associated pneumonia (VAP) and tracheal injuries
- Easy and intuitive operation
- For critical care, interhospital transport, and anesthesia
Ensuring optimal cuff pressure and patient saftey
Continuously optimized and controlled cuff pressure supports ventilation therapy and protects your patients from VAP and tracheal injuries – whether you use IntelliCuff during air transport with quickly changing ambient pressure, or in the operating room for N2O narcosis or laparoscopic abdominal surgery.
IntelliCuff is developed by Hamilton Medical and designed for immediate use; no calibration is required. It operates in a wide but still safe range of desired cuff pressures for various cuffed endotracheal tubes to provide suitable solutions for various clinical situations. You just set the desired cuff pressure, which is then maintained automatically.
Improved cuff pressure management
The leakage of oral secretions past the endotracheal tube (ETT) is a causative risk factor in the development of ventilator associated pneumonia (VAP), and excessive cuff pressure is a risk factor in tracheal injuries. Existing solutions for endotracheal tube cuff pressure management require manual monitoring and adjustment of cuff pressure, which is a critical aspect of the ICU staff workload. It has been shown that up to eight manual adjustments of cuff pressure are required daily to maintain recommended cuff pressure ranges (Sole 2011).
Recent clinical studies show that :
- Continuous cuff pressure control can decrease microaspiration and VAP (Lorente 2014, Nsei 2011).
- Continuous control is more effective in maintaining cuff pressure within an optimal range (Sole 2011).
观看我们在 YouTube 发布的视频，了解 IntelliCuff 的工作原理。
IntelliCuff pressure controller technical specifications
PDF / 147.5 KB
Cuff pressure management bibliography
PDF / 896 KB
Lorente L, Lecuona M, Jiménez A, Lorenzo L, Roca I, Cabrera J, Llanos C, Mora ML. Continuous endotracheal tube cuff pressure control system protects against ventilator-associated pneumonia. Crit Care. 2014 Apr 21;18(2):R77.
Nseir S, Zerimech F, Fournier C, Lubret R, Ramon P, Durocher A, Balduyck M. Continuous control of tracheal cuff pressure and microaspiration of gastric contents in critically ill patients. Am J Respir Crit Care Med. 2011 Nov 1;184(9):1041-7.
Sole ML, Su X, Talbert S, Penoyer DA, Kalita S, Jimenez E, Ludy JE, Bennett M. Evaluation of an intervention to maintain endotracheal tube cuff pressure within therapeutic range. Am J Crit Care. 2011 Mar;20(2):109-17.
Depending on your country, some features may be available as options, may have different specifications than described on this website, or may not be available. Contact your Hamilton Medical representative for details.